Types of pH SensorsDifferent pH sensors are made for diverse applications; hence, you need to understand how it works fully before purchasing any detector.
Combination pH SensorsThis is the most used type of sensor for a reason; it acts as the base for creating process sensors and laboratory sensors. In addition, its outfitted with two various electrodes, including a measuring electrode and reference electrode. The two sensors measure the tiny electrical difference within both electrodes.
Differential pH SensorsDifferential pH sensor is a heavy-duty sensor as it features an additional electrode, which helps prevent reference fouling. Unlike combination pH sensors, which have two sensors, differential pH sensors come equipped with three electrodes and work differently. The first two electrodes resemble the one in combination sensors, while the extra electrode is a metal ground conductor.
Digital pH meter tester
Laboratory pH SensorsThey are lightweight sensors that use combination pH sensor technology perfect for light applications, environmental sampling, and pool monitoring. One advantage is that they are customizable to fit the precise application expertise required. They come in three categories: basic, advanced, and research featuring pH1000, pH2000, and pH 3000, respectively.
Laboratory PH measurement
Process pH SensorsMade using higher combination sensor technology, process pH sensors have higher durability levels. Their basic sensors feature process connections, making them fit for pH levels continuous monitoring. You can mount them directly to a pipe or place them into a tank as the sensors have high durability.
Advantages and Disadvantages of pH Sensors
- It helps to determine the amount of primary and acidic any solution contains
- pH meters provide precise pH values since the pH meters are highly accurate with the help of pH sensors
- The meters offer numerical pH values directly
- Used in a wide variety of applications, including stain removal, pool maintenance, and cheese making
- it's the simple and fastest process of measuring pH
- You can calibrate it using the standard buffer solution
- Likelihood of breakage due to delicate glass electrodes
- Probability of pH calibrations due to carbon dioxide absorption and influence from temperature
- The pH often needs to calibrate
- Probability of process interruption due to deposits on the electrode membrane
- Calibrating it requires a particular buffer solution.
DIY PH Meter using PH Sensor & Arduino with OLED DisplayThis project will use a gravity pH sensor and Arduino to measure different aqueous solution pH and make Arduino pH Meter. Also, we'll learn how to calibrate the pH sensor to determine the sensor accuracy.
- Connecting wires
- I2C Module for LCD
- 1x Arduino Uno R3
- 16×2 LCD module with I2C
- Gravity Analog pH sensor kit
- BNC connector and gravity connector
Circuit DiagramThis circuit control board pattern will mount the component on a non-conductive board using a trivial alleyway known as traces. They will permit electrical components across the board to work by allowing the current to pass through.
SpecificationsThe pH sensor lit features two connected components: pH probe and transmitter V2 (signal conversion module). Their features include; Signal Conversion Module
- High accuracy measurement ±[email protected]℃
- Production current: 3.3~5.5V
- Detention range: 0~14
- Signal connector: PH2.0-3P
- Internal Resistance: <250MΩ
- Operating Temperature range: 5~60°C
- Neutral pint: 7±0.5
- Cable size: 100cm
- Electrode type: laboratory-grade
- pH Electrode Connection with Arduino
Arduino electronic platformArduino electronic board can read inputs, turn in into an output, light on a glass sensor, turn on an LED and activate a motor. The pH meter is made of glass and looks like a rod containing a "Glass membrane" tip. A buffer solution fills this membrane with pH =7, where the design of the electron warrants an H+ stable biding environment within the glass membrane. Once you dip the probe into the solution for testing, the hydrogen ions exchange with already positively charged ions. The charged ions are on the glass membrane creating an electrochemical potential forwarded to the electrical amplifier unit. As a result, they evaluate the potential between the two electrodes converting it to a pH module. Based on the Nernst equality, the difference between the two will determine the pH value.
Connecting pH Electron Conversion board with Arduino:Connect pH electron conversion board and Arduino as shown in the table below.
|Arduino||pH Electron Conversion board|
Encoding Arduino for pH MeterOnce you have connected the hardware successfully, you need to encode Arduino. First, include all the required libraries. When using the I2C LCD interface, including the "LiquidCrystal_I2C.h" library and Wire. H" for Arduino I2C functionality. While the calibration value is definite, it is still modifiable when necessary to give precise pH value of solutions. To smooth the output value, read ten simple analog values inside the loop(), and store them in an array. Afterward, categorize the received analog value in ascending order. It will help in the later stage when calculating the average of the running samples. To conclude, get the average values of the six center sample analog. Once you get the median value, it will convert into a definite pH value and print it on the LCD.
Calibrating pH Electrode
Calibrate equipmentIn this project, calibrating pH electrodes is very vital. What is required is a solution with a known value. For instance, it can be distilled water with a pH value of 7. Dip the electrode on the reference solution, and the LCD will display the pH value as 6.5. when calibrating it, take seven and subtract the Ph value, 6.5 (7-6.5=0.5). When coding, make the value 0.5+21.34=21.83. Once you have done with the modifications, upload the code to Arduino. Then dip the electrodes to the reference solution to recheck the pH. The correct pH value displayed on the LCD should be seven, but there could be a slight disparity, and it's acceptable. Likewise, calibrate the laboratory sensor by adjusting the calibrate and then checking all the solutions, ensuring you get the precise output.
Testing and ResultsAfter dipping pH electrons on different solutions, OLED will display will begin to show the pH value once you upload the code.
PH meter checking acidity level
Steps to Take to Ensure Accurate ReadingDespite the pH meter type you are using, there are steps you need to use to ensure you achieve an accurate reading;
- Ensure the signal conversion board and the BNC connectors are clean and dry. If they are dumb, they will interfere with the input impedance and inaccurate readings.
- Before you begin calibration, use a neutral buffer to test the pH meter.
- Please don't place the signal conversion board on a semiconductor or wet surface as it might interfere with the measurements. A nylon pillar is recommendable to fix the conversion board and allow a safer distance between the surface and the board.
- Ensure the sensitive pH probe head doesn't come into contact with any hard materials since any abrasion or damage will make the probe flop.
- Wash the electrode after use.
- Detach the pH meter from the signal board once you are done with the measurement. The signal conversion board might spoil if left in the power supply for an extended time.
Application of pH Sensors
Measure liquid fertilizer in a cup with digital PH meter neutral
- Seawater monitoring
- Beer and wine fermentation
- Food preservation and stability
- Natural water analysis
- Standard lab measurements
- Soil and mud analysis
- Blood analysis